Particle size composition test steps of laser particle sizer
1 Select "run" in the column of the tools Guizhou and Jiangsu in the station to publish the information of the inspection institutions of instruments and equipment related to epidemic prevention and control and the relevant detection capabilities to the public, and select "run cycle" in the drop-down menu of "run" to display the run cycle dialog box
2. In the operation cycle dialog box,
① select: check boxes for measurement compensation, alignment, background measurement, feed concentration measurement, input of sample information and input of operation information
③ select the pump speed (generally 70)
④ select ultrasonic period
3. Click the "sample information" button in the operation dialog box to enter the "input sample information" dialog box:
① you can enter the "group label" information, "sample label" information, notes, sample density, operation times, operator and other relevant contents in the dialog box as needed
fixture is the combination of some structures used by engineering plastics in many industries; ② Click the "selected liquid" button as needed, and the "selected liquid" dialog box will appear again. In the dialog box, select the sample test carrier options required for testing samples
③ click the "OK" button to exit the "selected liquid" dialog box
④ select the "OK" button again to exit the "input sample information" dialog box
4. Click the "operation information" button in the operation cycle dialog box to enter the "operation information" dialog box
① input as needed: running time, waiting time and running times
② select as needed; Check boxes for calculating granularity, storing files, outputting data, and printing reports
③ click the "directory" button as needed to select and store the files in the corresponding drive letter and folder
④ click the "sample table" button as needed to select the required optical model
⑤ click the "file name" button as needed, and select nine check boxes including "include ID", "include author" and "include date"
⑥ click the "output" button as needed, and select ten boxes such as "sample information", "Statistics", "average statistics"
⑦ click the "report" button as required, and select 15 boxes such as "sample information", "Statistics", "average statistics". Click the "OK" button to exit the "operation information" dialog box and return to the operation cycle dialog box.
5. Click the "OK" button in the operation cycle dialog box, the laser particle sizer price supply laser particle sizer starts to operate according to various instructions, and completes "measurement compensation", "alignment" and "measurement background" in turn. When the instrument runs to "measurement of feeding concentration", samples can be added to the sample table, Stop adding until the instrument prompts "OK", then select the "finish" button, and the instrument will start testing the sample until the sample test is completed.
6. After the sample test is completed, select "automatic cleaning" to complete the cleaning of the instrument
Copyright © 2011 JIN SHI